There will be 15~18 pcs test samples inside the temperature controlled chamber. Each test samples with 1~1.2 KG weight, Test samples without power on test, So there will not occur any additional heat or hot air during testing moment. According to customer’s testing requirement, We have made a water cooling solution for their testing purpose and they have accepted our proposal finally !
In the past few years, Imported temperature and humidity controlled chambers has always be first choice by the Chinese user. But in recent years, with the continuous development and innovation of refrigeration technique, domestic quality has changed the views of many Chinese users and gradually entered the world.
For example, HUAWEI mobile phone research and development has been recognized by the majority of users now, and it become the world’s top 500. The rapidly change heating and cooling stability chamber in the environmental test industry is also aimed at continuous innovation, and it will get better.
Chinese environmental controlled chamber, not inferior to foreign brands in terms of technology, such as csz chamber, thermotron chamber, tenney chamber. The main core system, High and low temperature devices use independent systems and mechanical single-stage refrigeration or cascade refrigeration, in which the commonly used cryogenic liquid nitrogen, liquid hydrogen, liquid helium.
In terms of cooling rate, the fastest Chinese environmental test chamber technology can reach 20℃/min. Chinese environment chamber has decades of low-temperature production experience, to provide users with stable, reliable environmental test equipment.
How fast can the cooling rate and heating rate achieve inside the testing room ? That is a good question. In fact, There are two methods to change the temperature rate of the environmental chambers: the nolinear movement and linear movement.
1. Nolinear Movement: Average rate of rise and drop in the whole process (that is, the rate of change in the whole process at N degrees), linear rate of rise and drop in the whole process (that is, the rate of change in N degrees per minute)
Average rate of temperature rise and drop in the whole process refers to the ratio of the difference between the highest temperature and the lowest temperature and the time within the range of temperature change in the small temperature controlled chamber. At present, the technical parameters of variable temperature rate provided by foreign manufacturers of temperature chambers refer to the nolinear movement.
2. Linear Movement: Linear rise and drop rate refers to the rate of temperature change that can be guaranteed in any 5 minute period, The temperature change rate will be guarantee.
In fact, for rapid temperature change temperature and humidity chamber, The most difficult part to guarantee the linear rising and cooling speed is the cooling rate that the climate chamber can achieve during the last 5 minutes of the cooling section of the vortex flowmeter. It is better for the climatic chamber have two parameters: the nolinear movement and linear movement functions. Generally speaking, The linear movement cooling speed is an average of nolinear movement cooling speed of 1/2.
In the case of abide by the laws of nature, in the long time use and transform the natural process, accumulate knowledge and skills and methods is a way people use natural transformation but need in the process of actual experience more can find their technology, rapid only continued innovation, litre drop in temperature test chamber can be done better.
Temperature heating and cooling test method for temperature controlled chamber: Temperature and cooling rate test for temperature control chamber should be carried out under the condition of full load. Temperature and cooling rate: the average rate of change for every 5 minutes should be 5±1℃/ min or humidity and temperature controlled chamber has the highest average rate of change.
The test point for temperature controlled environmental chamber is set as the geometric center point of the workspace. Wet bulb tester for climate test, accelerated test, hast test which is necessary and important to calculate the environmental chamber price.
What is the cooling and heating test program for temp controlled chamber ? Here are the guidance to make you understand how to do the temperature cooling rate and temp heating rate measurement work. We are not sure if the sub zero cincinnati environmental test chambers have such kind of comment, But we supply our customer these test program for their checking purpose.
1 Within the adjustable range of temperature of the temp chamber, the lowest nominal temperature is selected as the lowest cooling temperature, and the highest nominal temperature is the highest heating temperature.
2 Turn on the power, make the temperature controlled vacuum chamber from room temperature to lowest temperature, stable at least 3 hours, rise to the highest temperature, stable at least 3 hours later to the lowest temperature. During heating and cooling period, record once per minute until the end of the test process.
|3 Calculation and evaluation of test results|
4 Methods for checking and evaluating the sealing performance of refrigeration systems. Halogen lamp or soapy water shall be used to check the sealing condition of the refrigeration system pipe joints. If there is no leakage, humidity test chambers shall meet the requirements of section 4.2.3
|5 Methods for testing and evaluating thermal insulation performance|