When you bought an environmental chamber from Wewon Environmental Chambers Co., Ltd. Please make sure you will do the clean work by yourself or your service engineer.
If you want extend the environmental chamber’s lifetime, and make sure the environmental chamber with a good testing performance. Clean your environmental chamber and remove the dirty and dust from this chamber is so important.
Below is the guidance to teach you how to do the clean work for your environmental chamber unit. Simply to say, If you made your environmental chamber with clean surface and no oil dirt, no dust to stop the water pipes, Then your environmental chamber will work well and has no any quality problem happen during your testing moment.
|Daily Clean Work and Maintenance Notice:|
|Spare Parts Description:||Situation and Solution:||Maintenance Time:|
|①||Wet Bulb Gauze Replace Work|
|②||Check the Water Level (Wet bulb gauze water tank, Humidification barrel, Bottom water tank in drawer)||Frequency or Regular Intervals||Every 3 Months|
|③||Cleaning the Condenser|
|④||Clean the Testing Room|
|⑤||Clean the Body Structure||Clean it at Regular Intervals||Every 4 Months|
|⑥||Clean Power Electronic Circuit Part||Clean it at Regular Intervals||Every 4 Months|
|⑦||Clearn Water Tank in Drawer||Clean it at Regular Intervals||Every 1 Months|
|⑧||Clean the Water Out Filter||Clean it at Regular Intervals||Every 1 Months|
|⑨||Clean the Water in Filter||Clean it at Regular Intervals||Every 1 Months|
|●||Do not start frequently, Power supply voltage, use environment, ventilation, The internal impurities should be removed before operation.|
|●||In summer, When the indoor environment is too high, over-current alarms are prone to occur. It is recommended that the use environment be kept below 25 degrees so as not to affect normal use.|
|●||After each test, the temperature and humidity test chamber should be kept clean to prevent corrosion.|
|●||The control testing room of the constant temperature and humidity test chamber is cleaned once a month to prevent excessive dust and affect the normal operation of the equipment.|
|●||After the end of the experiment, the main power supply should be turned off, the water traces in the workroom should be cleared, the test room should be kept in a boring environment, and the appearance should be kept neat.|
|●||If the temperature of an electrical component is too high, it is necessary to immediately check the cause of the attack.|
|●||The electrical components on the control panel of the constant temperature and humidity test chamber shall not be dismantled if the malfunctions need to be exchanged, so as to ensure the service life of the test room.|
|●||After long-term use of equipment such as motor blower equipment and turntable deceleration equipment, the constant temperature and humidity test chamber should be checked and watch for clean fuel if an abnormal noise occurs.|
|●||The power distribution room of the constant temperature and humidity test chamber shall be cleaned at least once a year or more. When cleaning, vacuum cleaners can be used to remove indoor dust. The outside of the chamber must also be cleaned more than once a year.|
Meanwhile, Please make sure your testing laboratory room Temperature Range: +15C ~ +25C, Treat your environmental chamber with a good weather temperature and the quality problem will never come to find you. Enjoy your testing work !
The below mentioned model is the popular design and be ordered frequently. If you have the new inquiry for environmental chamber, Please inquiry to us timely. We accept the custom design. Welcome the engineer share the ideas with us, And we will make your demand design to be truth. Thank you.
|●||Model:||TH-0225L||●||Temp Range:||-40C +150C||●||Humidity Range:||20%~95%|
|●||Model:||TH-0408L||●||Temp Range:||-70C +150C||●||Humidity Range:||20%~98%|
|●||Model:||TH-0800L||●||Temp Range:||-40C +150C||●||Humidity Range:||20%~95%|
|●||Model:||TH-1000L||●||Temp Range:||-70C +150C||●||Humidity Range:||20%~98%|