The salt spray chamber specification and technical parameters is important for a new engineer who has no experience for a salt spray chamber operation work. We have an Austria customer, He asked the salt spray chamber specification and technical parameters from our technical dept. Because he has no idea on his testing process.
Below are some salt spray chamber specification according to our Austria customer mentioned. Let me do a summary here to make the new engineer easier understand the salt spray chamber specification. It should be a convenient way to improve the new engineer operation skills for the salt spray chambers.
Salt Spray Chamber Specification And Technical Parameters｜The salt spray chamber simulates corrosion test in artificial environment. It is a simple and effective method to check whether metal materials or other test item has corrosion protection or not. It will be suitable for corrosion test of discontinuity, control and damage in organic and inorganic coatings.
Meanwhile, For a quality control purposes, Comparisons can be made between test samples coated with the same paint. Salt spray tests are particularly useful for detecting discontinuities, proportionally small holes and other defects in certain metals, organic, phase-grade oxidation and conversion coatings.
|Question:||When I processed a salt fog test according to ASTM B117, The water in the test room and salt water tank will choose distilled water or normal water ?|
|Question:||After the first test runs and the first cleaning we found some red residues (rust?) in the test chamber and in the sewage of the waste water outlet – Have you ever seen this problem ? What is the root cause of this rust ? Is there any bad influence to the testing results ?|
|Answers:||This is the normal circumstances. Because the salt spray corrosion is the maximum damage when you processed an environmental test. After a fully salt and corrosion test, The surface of PVC plate may occur some rust.|
|Question:||How often should we change the water in the test room and salt water tank ? Manual water export ?|
|Question:||How often and how should we clean the salt spray chamber ? When I checked the salt spray chamber specification user manual, I can not find the explanation there.|
|Question:||Is it necessary that two measuring cups are closed with this white pipe plug ? If not what will be the consequence – higher PH-Value ?|
|Answers:||Equip the white cup on this two measuring cups are not compulsive. Make this white cup on or not on, Will no affect your testing result. The salt spray chamber specification and technical parameters user manual has some guidance to teach you how to equip the measuring cups. Meanwhile, We will make some short videos to guidance customer how to process their testing work.|
|Question:||Two Measuring Cups｜How and how often should we check the PH-Value of this water ?|
|Question:||Two Measuring Cups (50 ml)｜Could we use bigger cups (up to 100 ml)? During the weekend we can’t pour out these cups, so they will overflow.|
|Question:||Two Measuring Cups (50 ml)｜If there is a difference in the amount of water between the two cups – What can we do/ adjust that the amount will be almost equal ? Does this mentioned on your salt spray chamber specification user manual ?|
|Answers:||Please no worries about this issue. During the testing, The salt fog occur. The left side collection tube and right side collection tube may receive the salt fog has some difference. It’s normal circumstances.|
In fact, These technical questions comes from our Austria customer. A new engineer use our salt spray chamber and it seems he has no experience to use the salt spray chamber for his test before. We have prepared some salt spray chamber specification and answer the customer questions timely when we are available.
|Question:||Where can I find a salt spray corrosion test chamber price ? The Wewon salt spray corrosion test chamber price will be cheaper than Ascott salt spray chamber price ?|
|Answers:||We consider that the Ascott salt spray chamber quality is good. Because they are the world famous and supply the salt spray chambers to different customers many years. The Wewon salt spray chamber price should be cheaper. If you want to save the salt spray test chamber cost, Buy a Wewon salt spray chamber should be a good idea.|
|Question:||What is the salt spray chamber hsn code ? Where can I do the salt spray chamber calibration ? What is the salt spray chamber working principle ?|
|Answers:||In China, We mostly use the HS code for salt spray chamber export work. Salt spray chamber HS code what we frequently used: 8424899990. Regarding the calibration, We have a supplier which is a independent third party certifier provider. We will pay the cost and request them come for a calibration work before shipment.|
|Question:||We prefer the salt spray chamber manufacturers in Delhi, in bangalore India. Can you support this ? Do you have the sales representative in Chennai India ?|
|Answers:||For this moment, We have some sales order from Delhi and bangalore India. If you need the local installation work, We prefer to find a local service engineer to support you. We will pay the service cost to the engineer who worked for us.|